- UL®1069 Listed
- All stations interchangeable
- Simplified multiplex wiring and plug-in connectors
- Input available for Hill-Rom® Sidecom® beds or remotecall-in button applications
- Uses pillow speakers or push-button call cords
The IR319B Single Patient Station provides reliable twoway, hands-free communication between patient and nurse, and provides for programmable privacy, paging and patient priority from the master station. Calling the nurse is accomplished by pressing the call cord button or the area indicated nurse on the pillow speaker. Solid state LEDs on the panel indicate the call has been placed and the intercom is in use.
If the pillow speaker or call cord is accidentally removed from its receptacle, the nurse master station registers a non-cancellable cord out call. The call may be reset only by returning the call cord or pillow speaker to its receptacle.
The IR320B Dual Patient Station has the same specifications and functions as the IR319B, plus an additional independently operating call cord/pillow speaker receptacle and a second call-placed indicator.
Name : TekTone IR319B Single Bed Station with DIN Jack , TekTone
Category : Patient Stations
UPC Code : 812664020351
Country of Origin : United States. Country of origin is subject to change.
ADI Part Number : T9-IR319B
Brand Name : TekTone
Manufacturer : TekTone Sound & Signal Mfg., Inc
Manufacturer Part Number : IR319B
Product Line : Tek-Care
Product Name : Nurse Call Patient Station
Product Type : Nurse Call Patient Station
Application/Usage : Emergency, Hospital, Indoor, Intercom System
Depth : 2.8"
Depth (mm) : 69.85
Form Factor : Surface-mountable
Height : 4.5"
Height (mm) : 114.3
Product Material : Plastic
Width : 6.4"
Width (mm) : 161.925
Compatibility : TekTone Tek-CARE 300II Nurse Call System